Guangzhou Xinghe Aluminum Composite Panel Co., Ltd.
We provide customers with quality products and provide high-quality services
If you would like to leave us a comment please go to contact us
Have you ever used aluminum plastic composite panel ? Aluminum composite panel manufacturer-Xinghe, specializes in the field of brushed aluminum composite panel, fire-proof aluminum composite panel, marble aluminum composite panel and so on.
We provide our customers with more effective knowledge and products. Maybe you have some questions about aluminum plastic composite panel’s details. Let me tell you some of them.
1.Ambient temperature and relative humidity: (23 ± 5) ° C, (50 ± 20)%;
2.Power supply voltage: 220V±10%, 50HZ;
4.gas source: purity ≥ 95% propane;
5.the instrument dimensions: about 840mm × 400mm × 810mm;
6. The Bunsen burner can apply flame time to the sample for 15s and 30s;
7.Aluminum-plastic panel flammability test furnace sample size mm: length 250-1mm, width 90-1mm, thickness ≤ 60mm.
1. Check if the “Ignition Time” display setting on the instrument panel is 15s or 30s, and the “Test Time” display setting is at the maximum range;
2. Turn on the power supply and air supply, close the “gas switch” valve on the panel, and open the cylinder valve;
3. The flammability test furnace of the aluminum-plastic panel opens the “power switch” of the flammability test furnace of the building materials, and presses the “reset” button to reset the burner;
4. Surface ignited sample: Mark the marked line at 40mm and 190mm on one long side of the sample, and install the side of the scribe line down on the sample holder, fix it in the combustion box, and tilt the burner by 45°. Press the “Run” button to adjust the distance from the burner port to the sample so that the flame is applied to the 40mm mark at the center of the pattern:
(1) Open the “Gas Switch” on the flammability test furnace of the building material. Press the “Ignition” button to point the burner. Adjust the height of the knob on the burner to (50±1) mm and tilt the handle 45° to close the door.
(2) Press the “Test” button to apply a flame to the sample for 15 seconds. After the flame is applied, the burner is automatically removed. The time from ignition to the time when the flame reaches the marking line or the surface of the sample is extinguished is extinguished. When the flame is extinguished, Press the “OK” button, the “Test Time” display on the instrument panel stops and the data is recorded.
5.Edge-ignited sample: mark the line with a length of 150mm on one side of the sample, and install the side of the line down on the sample holder, and fix it in the combustion box so that the burner flame is applied to the surface after 1.5. The center position of the bottom edge width of the mm sample.
(1)Click on the burner. Press the “Test” button on the control panel to apply flame to the sample by the burner flame. After the flame is applied, the burner is automatically removed. The measurement starts from the ignition until the flame reaches the marking line or the surface of the sample burns. When the flame is off, press the “OK” button, the “Test Time” display on the instrument panel stops, and record this data.
(2)After repeating 6 samples, turn off the power supply, “gas switch” and cylinder valve, and perform necessary maintenance on the combustion box and sample holder.
1.Sample holder and 0.5A fuse each 1
2.Power cord 1
3.Rubber tube 1
4.Instruction manual 1 co
5.Product certificate 1 copy
After reading this blog, do you want to learn more about us? Please contact us for further information.